Traceable Humidity/Temperature Meter
Digital display permits instant readings of relative humidity and temperature (°F/°C). Output jack provides a 10 millivolt signal per degree or per 1% relative humidity. Use with a chart recorder, strip recorder, or computer that accepts millivolt signals. It is ideal to monitor conditions in chambers, clean rooms, food processing, scientific labs, and all humidity-sensitive manufacturing operations. Recorder output provides a hard-copy record.
Instantly updates readings
Precision thin-film capacitance sensor updates the display two-and-a-half times per second. Large, bright, 1/2-inch-high four-digit liquid crystal display can be read from 10 feet away. Rugged handheld unit is completely portable and designed for years of dependable and reliable service.
Traceable to NIST:
To assure accuracy an individually serial-numbered Traceable® Certificate is provided from an ISO 17025 calibration laboratory accredited by A2LA. It indicates traceability to standards provided by NIST (National Institute of Standards and Technology).
- Part number COCSEN069
- Features a recorder output
- Features a HOLD function
- Humidity range is 10.0 to 95.0%RH
- Temperature range is 4.0 to 140.0°F (20.0 to 60.0°C)
- Resolution is 0.1° for both readings
- Accuracy is ±2.5%RH mid-range to ±5%RH elsewhere
- Response time is 30 seconds to 3.5 minutes
- Size is 2-1/2 x 10-1/2 x 1 inches
- Weight is 7 ounces
- Supplied complete with Traceable Certificate 9-volt alkaline battery and carrying case
- Handheld Temperature Humidity Meter
- 9V Battery
- Hard carrying case
- Traceable® Certificate
- Expert Technical Support
Low battery power can occasionally cause any number of “apparent” operational difficulties. Replacing the battery with a new fresh battery will solve most difficulties.
An erratic display, faint display, no display, or battery symbol appearing on the display are all indicators that the battery needs replacement.
Slide the battery cover off the back of the meter in the direction of the arrow. Remove the exhausted battery and replace it with a new 9-volt alkaline battery.
Replace the battery cover. The data logger feature will store and retain up to 99 readings until they have been cleared.
The number at the bottom left of the display indicates how many readings are currently stored in the data logger memory.
If desired, prior to recording additional readings into memory, clear the existing data logger memory. (See the “Clearing the Data Logger Memory” section.)
With the meter turned on and displaying the desired values, press the MEMORY button.
Each time the MEMORY button is pressed, “M” will appear to the left of the display and the reading will be recorded in the data logger memory.